The Pocket Colorimeter II™ has two manuals. The Basic and Full User Manuals can be found below. All Pocket Colorimeter™ II instruments. This symbol, if noted on the instrument, references the instruction manual Hach Pocket Colorimeter™ instruments* are low-cost, high-quality filter photometers. Consult HACH’s Pocket Colorimeter II brochure on ArchiExpo. Page: 1/4.
|Published (Last):||12 June 2018|
|PDF File Size:||11.23 Mb|
|ePub File Size:||19.32 Mb|
|Price:||Free* [*Free Regsitration Required]|
May not develop full color or color may fade instantly. Use of the factory calibration is recommended unless the user is required to generate a calibration. Enter text from picture: Failure to do so could result in serious injury to the operator or damage to the equipment.
To print the manual completely, please, download it. The Pocket Colorimeter II is designed to measure solutions contained in sample cells.
No manual calibration is ever required. Move the Pocket Colorimeter instrument away from the device receiving the interference.
Each data pair requires a concentration value and the absorbance value for the given concentration. Simply zero the instrument with a blank, insert the reacted sample, and read the result.
Obtain a new set of standards, or use the Pocket Colorimeter II to assign new values to existing standards. The instrument has two channels in which measurements can be made.
Operation is subject to the following two conditions: Printed in the U. Hach Instruction Manual pages.
Hach POCKET COLORIMETER II Instruction Manual
Each instrument comes in a ready-to-use kit that includes a manual, sample cells, and a sturdy custom carrying case. Open the catalog to page 1. Precautionary Labels Please pay particular attention to labels and tags attached to the instrument. For example, if five points have been entered, hafh can be deleted using this feature.
Hach Pocket Colorimeter™ II for Chlorine or Manganese | Pine Environmental
Don’t show me this message again. Many are based on EPA approved methods. Forums Documents and Manuals. Press the key twice to exit and accept the changes. Determine amount to be added on a separate mL sample, then add the same amount to the sample being tested. No need to carry pen and paper to manually record results. Record 10 most recent data points along with cklorimeter of measurement. A coloeimeter, impact-resistant shell protects the electronics and optics, while a lanyard keeps the cap attached to the colorimeter.
Single Parameter go anywhere portable photometer Lightweight and battery operated, suitable for extended field work or quick, on-the-spot process monitoring.
Order information | Hach-Lange POCKET Colorimeter II User Manual | Page 4 / 4 | Original mode
Weighing in at only. Press and hold again to turn off the backlight.
Simply zero the instrument with a blank, insert the reacted sample, and read the result. Page 88 On the basis of strict liability or under any other legal theory, in no event shall Hach Company be liable for any incidental or consequential damages of any kind for breach of warranty or negligence.
Chlorine Powder Pillow to Do not use the same Note Menu Selections Press the key to access the menu selections. Tightly cover the cell with the instrument cap.